Measuring the activity of BioBrick promoters using an in vivo reference standard
© Kelly et al; licensee BioMed Central Ltd. 2009
Received: 04 May 2008
Accepted: 20 March 2009
Published: 20 March 2009
The engineering of many-component, synthetic biological systems is being made easier by the development of collections of reusable, standard biological parts. However, the complexity of biology makes it difficult to predict the extent to which such efforts will succeed. As a first practical example, the Registry of Standard Biological Parts started at MIT now maintains and distributes thousands of BioBrick™ standard biological parts. However, BioBrick parts are only standardized in terms of how individual parts are physically assembled into multi-component systems, and most parts remain uncharacterized. Standardized tools, techniques, and units of measurement are needed to facilitate the characterization and reuse of parts by independent researchers across many laboratories.
We found that the absolute activity of BioBrick promoters varies across experimental conditions and measurement instruments. We choose one promoter (BBa_J23101) to serve as an in vivo reference standard for promoter activity. We demonstrated that, by measuring the activity of promoters relative to BBa_J23101, we could reduce variation in reported promoter activity due to differences in test conditions and measurement instruments by ~50%. We defined a Relative Promoter Unit (RPU) in order to report promoter characterization data in compatible units and developed a measurement kit so that researchers might more easily adopt RPU as a standard unit for reporting promoter activity. We distributed a set of test promoters to multiple labs and found good agreement in the reported relative activities of promoters so measured. We also characterized the relative activities of a reference collection of BioBrick promoters in order to further support adoption of RPU-based measurement standards.
Relative activity measurements based on an in vivoreference standard enables improved measurement of promoter activity given variation in measurement conditions and instruments. These improvements are sufficient to begin to support the measurement of promoter activities across many laboratories. Additional in vivo reference standards for other types of biological functions would seem likely to have similar utility, and could thus improve research on the design, production, and reuse of standard biological parts.
The engineering of many-component, synthetic biological systems is being made easier by the development of collections of reusable, standard biological parts [1–6]. Standardization of components has been instrumental in managing complexity in other engineering fields by helping engineers to reliably design and deploy systems comprised of combinations of parts . However, it is an open question whether the overwhelming complexity of living systems will prevent biological engineers from fully achieving similar capabilities (below). To help answer this question, a Registry of Standard Biological Parts started at MIT now maintains and distributes thousands of BioBrick standard biological parts . BioBrick parts provide the first popular example of standard biological parts. However, BioBrick parts are currently only standardized in terms of how individual parts are assembled into multi-component systems (that is, "physical composition") [1, 9].
The utility of so-called standard biological parts would increase if the behavior of parts, both in isolation and in combination, were more predictable (that is, "functional composition"). Prediction of behavior, in turn, depends on the initial designs and refinement of the parts themselves, the characterization of part functions, and the representation of part functions via abstract models (for related examples see [10–12]). Today, most BioBrick parts are directly derived from natural DNA sequences with only slight modifications to support at least one physical assembly standard, and many parts remain to be characterized. For example, fewer than 50 out of over 500 transcriptional promoters now available via the Registry have been characterized. Making matters worse, for the 50 characterized promoters, the methods of characterization are disparate and the resulting data incomparable. Shared and standardized approaches are needed in order to begin to address the challenge of characterizing promoters (and other types of standard biological parts) across a distributed community of biological engineers.
Making reliable and comparable in vivomeasurements of biological parts has proven challenging. For example, five different efforts to measure the abundances of proteins in the yeast pheromone mating response system, one of the best characterized eukaryotic signalling systems, produced reports for the numbers per cell (abundances) of key system proteins that vary over a factor of ~12 [Thomson TM, Benjamin KR, Bush A, Love T, Pincus D, Resnekov O, Yu R, Gordon A, Colman-Lerner A, Endy D, Brent R: Scaffold number in yeast signaling system sets tradeoff between system output and dynamic range. Molecular Systems Biology. Unpublished]. Such examples suggest that measurement of the state or activity of biological systems, whether natural or engineered, may be unlike past engineering experiences, in that the minor differences in experimental conditions (relative to what can be readily controlled for, below) may cause large changes in the properties being measured. Even if conditions could be controlled for, it has proven challenging for researchers to develop and adopt standard approaches for characterizing biological parts. For example, an analysis of 80 published papers in which researchers used beta-galactosidase (β-gal) activity as a measure of gene expression found that at least six different protocols were used to measure enzyme activity . In addition, nearly all activities were reported in "Miller units" even though in several cases there were differences in the substrates used to quantify enzymatic activity (CPRG or ONPG), the experimental conditions (pH and temperature for the assay), and even the absolute units of the Miller unit (nmol/min or μ mol/min) . Differences in conditions such as using either CPRG or ONPG as a substrate for enzymatic assays lead to incompatible results , and thus Miller units should generally not be considered comparable unless they have been calibrated against a common reference standard .
The challenge of making reliable in vivomeasurements of biological parts is further compounded by the need to measure many part properties indirectly via biological "measurement instruments" such as reporter proteins whose production can also be sensitive to experimental conditions. For example, β-gal activity can be used as an indirect measure of the behavior of a promoter, but the translation and activity of the β-gal protein is itself sensitive to experimental conditions such as temperature or choice of media. Since both the measurement instrument (β-gal) and the property being measured (promoter activity) are sensitive to measurement conditions (perhaps in differing ways) correcting for errors in measured promoter activity due to changes in conditions is more difficult. In theory such challenges could be addressed by strict adherence to standard measurement conditions. However, the adoption of standard measurement conditions in biological engineering is prevented by both practical constraints (as noted above) and also engineering constraints, such as culture or performance requirements that are specific to a particular biotechnology application. The overall situation is summed up nicely via the following quote: "There is no such thing as a standard (biological) component, because even a standard component works differently depending on the environment" .
Although the characterization of standard biological parts is challenging, lessons from the measurement of other types of physical objects are worth considering. For example, one approach to controlling for variation in the measured property of an object in response to changing experimental conditions is to collect data from which to develop a model that describes the relationship between the measured property and experimental conditions. As a specific example, models based on empirically determined coefficients of thermal expansion for common building materials (for example, Oak = 54E-6/K at 20°C; Stainless Steel = 17.3E-6/K at 20°C) are now sufficient to enable the reliable construction of structures across a range of environments . However, given the complexity of living matter, the relationships between the measured properties of biological parts and experimental conditions may be difficult to determine (at first). Thus, a second lesson worth considering is the measurement of relative (or ratio) properties rather than absolute characteristics. A relative measure is the ratio of the measurement of some aspect of the object being characterized in comparison to a standard reference object that is measured under the same conditions. For example, early methods for the diagnosis of osteoporosis made use of a measure of spinal cord deformity that was based on the ratio of various length measurements of vertebra within an individual patient . Doctors, by using a relative measurement for length, could account for variation in vertebra sizes between individuals of different body types or heights. As a second example, microarray experiments are frequently performed by co-hybridizing probes synthesized from both a reference and experimental RNA sample that have been labelled with different colors ; gene expression levels are then reported as the ratio of the experimental and reference intensities on each array spot. Thus, measurements made in relation to defined reference standards may provide an important first approach in characterizing the in vivoactivity of biological parts and, over time, could enable the collection of empirical data sufficient to support the development of models that describe the effect of varying conditions on part properties.
Here, we characterized the in vivoactivity of BioBrick promoters in order to evaluate if measuring relative activities might provide a useful initial framework for measuring the activity of standard biological parts across varying conditions. We chose to characterize promoters as a first example since they are ubiquitous in engineered biological systems, relatively well-understood, practically useful to biological engineers, and poorly characterized in the existing BioBrick collection [20, 21]. We developed a system that allows indirect measurement of the activity of promoters via observation of the synthesis rate of Green Fluorescent Protein (GFP) encoded by mRNA transcribed from each promoter. Our system requires the use of a quantitative model that allows promoter activity to be estimated from observed rates of GFP synthesis (below). Using this approach we demonstrate that normalizing the apparent absolute activity of a promoter to a defined reference standard promoter can help account for variation in conditions that would otherwise lead to significant differences in reported measurements.
Definitions and models for absolute promoter activity
Our first step in characterizing standard biological promoters was to choose the property or properties whose measure would best support the reuse of such parts by biological engineers. Since the primary use of promoters is to initiate transcription, we chose the rate of transcription initiation as the property to be measured. We next chose the promoter clearance rate as the specific property that best describes transcription initiation; we refer to this property as "promoter activity" throughout. In turn, we defined promoter activity as the number of RNA polymerase molecules that pass by (or clear) the final base pair of the promoter and continue along DNA as an elongation complex. We report promoter activity using the generic unit of "Polymerases Per Second," or PoPS, in place of the more traditional "promoter clearance rate" because reporting activity in PoPS allows promoters to be directly compared to other genetic parts whose functioning impacts elongating polymerases, such as transcription terminators . Other properties of promoters such as the binding constant of RNA polymerase to the DNA encoding the promoter, or the secondary structure of the DNA were not considered; while such properties may be relevant to researchers who are studying or engineering new promoters, our focus here was to support researchers who are characterizing or reusing existing promoters.
Directly measuring PoPS in vivois challenging and, to our knowledge, has not yet been reported. However, by placing a promoter upstream of the coding sequence for green fluorescent protein (GFP) we could use the rate of GFP synthesis as an indirect measure of promoter activity. We could then use a quantitative model to relate observed GFP synthesis rates to promoter activities reported as PoPS.
We adopted a previously described ordinary differential equation (ODE) model of GFP expression from a constitutive promoter to relate GFP synthesis rates per cell to promoter activities [9, 23]. We evaluated this ODE model at steady-state (Additional file 1) in order to determine the rates of successful mRNA initiation events per DNA copy of each promoter (PoPS SS ) given observed GFP synthesis rates per cell ( ):
where γ M is the mRNA degradation rate, a is the GFP maturation rate, γ I is the degradation rate of immature GFP, ρ is the translation rate of immature GFP from mRNA, and n is the number of copies of the promoter in the cell.
Variability due to equipment and conditions
Definition and models for relative promoter activities
We noted that the activity of promoters (for example, J23101 and J23150) measured across different conditions or with different instruments was correlated (Figure 1A). This correlation suggested that a measure of relative promoter activity might be less sensitive to varying experimental conditions or measurement instruments. To test this idea, we defined a new property – relative promoter activity – as the ratio of the absolute activity of a sample promoter, φ, relative to the absolute activity of a standard reference promoter, BBa_J23101, with both promoters measured under equivalent conditions and with the same measurement instrument. We reported relative promoter activities in a newly defined unit: Relative Promoter Units or RPUs. By our definition, a sample promoter with a relative activity of 1 RPU has activity equivalent to BBa_J23101.
Taken together, by reporting promoter activity relative to a reference standard promoter (BBa_J23101) and choosing promoters with identical transcription initiation sites and identical sequences downstream of the initiation sites, researchers can quickly report measured relative promoter activities in compatible units without having to independently measure GFP maturation rates, mRNA degradation rates, protein production rates, or plasmid copy number for their specific experimental setup. (We detail the precise numerical sensitivity of the quantitative model to each of the above assumptions in Additional file 1).
We converted GFP synthesis rates measured across 7 different conditions and instruments (Figure 1A; Coefficient of variation (CV) of the measurements is 39.1%) to relative promoter activity in RPUs (Figure 1B; CV of the measurements is 17.5%). We noted that the coefficient of variation in promoter activity was reduced by approximately half when converted to RPUs from GFP synthesis rates. This reduction in variation suggests that relative promoter activity might be a useful property for characterizing promoters. However, care should be taken to note that while relative promoter activities remain fairly constant across some range of conditions, absolute promoter activities vary widely across these same conditions. Stated differently, a promoter that has an equivalent relative activity across multiple conditions might not produce equal absolute activity (as measured in PoPS) across the same conditions (please see Discussion).
Community-based measurement of promoter collections
To further support community-based standardized measurement of promoter activities, we developed a first generation measurement kit for characterizing the relative activity of BioBrick promoters in RPUs. Our overarching objective for the kits was to enable independent researchers to make comparable measurements of relative promoter activity in standard units. We developed instructions and a parts list for the promoter measurement kit (Additional file 1 [Supp Box1 and Supp Table1]). The promoter measurement kit contains measurement "instruments" such as a green fluorescent protein (GFP) reporter device (BBa_E0240) and backbone plasmid (pSB3K3), as well as a recommended E. coli strain (TOP10). The reference promoter (BBa_J23101) was inserted upstream of the GFP reporter device (BBa_E0240) and included in the kit as the reference standard construct (BBa_I20260). In order to measure the activity of a user-specified promoter, kit users assemble the user-specified promoter upstream of the GFP reporter device and insert this combined part into the backbone plasmid to form the promoter test construct. The process for inserting a promoter upstream of the GFP reporter device is based on three-antibiotic BioBrick standard assembly , and is outlined in the instructions included with the kit (Additional file 1 [Supplementary Box 1]).
We found the absolute activity of promoters to vary under different experimental conditions and when using different measurement instruments. We chose a promoter (BBa_J23101) to serve as a reference standard and demonstrated that by measuring relative promoter activity (activity of a sample promoter divided by activity of the reference standard promoter BBa_J23101, measured under the same conditions) we could reduce reported variation in measured promoter activity across differing experimental conditions and equipment. We defined the Relative Promoter Unit (RPU) in order to enable researchers to report promoter characterization results in compatible units, and developed a measurement kit in order to more easily allow researchers to adopt an RPU-based measurement approach. We distributed a test set of 4 promoters to 7 independent labs and found good agreement in the measured relative promoter activities across the test set. Finally, we characterized the relative promoter activity of 7 BioBrick promoters in order to bootstrap a collection of promoters measured according to our initial RPU reference standard.
Absolute and relative promoter activities
The absolute activity of a promoter is defined by the number of elongating polymerases per second (PoPS) exiting the promoter. The same promoter under different environmental conditions can have widely varying absolute promoter activities (Figure 1A). Moreover, it is challenging to relate an indirect measure of absolute promoter activity, such as per cell GFP synthesis rates, to PoPS since any variation across conditions in the functioning of the genetic measurement instrument may not be well correlated with variation in promoter activity across these same conditions. In contrast, the relative activity of a promoter is defined as the ratio of the absolute activity of the promoter to the absolute activity of a reference standard promoter measured under the same conditions and with the same measurement instrument. We found that the relative activity of a promoter will remain fairly constant across a practical range of conditions (Figure 1B).
Relative promoter activity as reported in RPUs enables the ranking of the activities of promoters but does not, by itself, provide information about the absolute activity of the promoter under particular conditions. For example, if the relative activity of a sample promoter remains constant across several conditions, it is not necessarily the case that the promoter is producing equal numbers of mRNA transcripts in each condition, only that the promoter activity is remaining proportional to the reference standard across the conditions. Nonetheless, relative promoter activity is a valuable property to measure and report since promoters can be rank ordered by relative promoter activity even if they were characterized by different researchers or across different environments (Additional file 1 [Supplementary Table 4]). Furthermore, if both the relative and absolute activities of a promoter are measured, then a conversion factor can be established that defines the relationship between relative and absolute promoter activity under specific measurement conditions. For example, the reference standard promoter J23101, which has a relative activity of 1 RPU, has an estimated absolute activity of ~0.03 PoPS under specific conditions (Additional file 1). Therefore, the promoter J23151, which has a relative activity of approximately 0.5 RPU under these same conditions, would be predicted to have an absolute activity of ~0.015 PoPS. As more absolute measurements are made, an expanded set of conversion factors (or functions) could be developed, allowing for improved estimates of absolute promoter activities across a wider range of measurement conditions.
Interlaboratory measurement of promoter activity
Measurement of in vivopromoter activities across laboratories is challenging due to the sensitivity of results to both experimental conditions and measurement instruments (Figure 1), as well as the lack of shared reference standards . We demonstrated here that by using a relative promoter activity to characterize promoter strength based on a shared reference standard, seven independent laboratories could make comparable measurements of three promoters (Figure 2). We expect that future improvements to the recommended measurement techniques and measurement kit components could further reduce the variation in measurements across laboratories, with the results reported here providing a practical baseline for judging proposed improvements.
The reference promoter BBa_J23101 and relative promoter activity measured in RPUs provide a shared platform for researchers to evaluate different measurement procedures. We deliberately have not advocated a single measurement procedure, and there should be many acceptable procedures for characterizing promoters in RPUs, just as units for length or mass are not tied to a single measurement approach. The choice of the best measurement procedure will be influenced by the particular group making the measurements. For example, laboratories without access to equipment for capturing high-throughput single-cell measurements of fluorescence might opt for a bulk fluorescence measurement using a fluorimeter. Other groups might prefer to obtain single-cell measurements using quantitative microscopy or flow cytometry, or to capture a time-course of fluorescence measurements from a growing culture. As different measurement procedures are likely to have merits within different communities we expect that a number of procedures will be established. As an example, for the community of undergraduate teams using the promoter measurement kit during the International Genetically Engineered Machines (iGEM) competition, we have suggested a measurement protocol that can be easily carried out by novice researchers and that only requires two absorbance and bulk fluorescence measurements (Additional file 1 [Supplementary Box 2]) .
Engineering with characterized promoters
We anticipate that both absolute and relative promoter measurements will be useful in engineering genetic networks, however it is unclear to what extent one approach might be preferred over the other (presuming both types of measurements could be readily obtained). For example, we can imagine an engineering design framework in which the absolute activities of promoters and other functional genetic elements are tracked explicitly, in order to support detailed modeling and analysis of issues such as the absolute "load" placed on a host cell via recombinant gene expression. Such an ability seems likely to become more important as many-component engineered biological systems are attempted (dozens to hundreds of gene products), in which the absolute expression levels of individual genes must be well managed and might be kept low compared to the high-expression, protein production systems typically used today. However, we can also imagine a competing or complementary engineering framework, based on the idea that cells already provide self-adapting and robust environments within which the absolute activities of genetic elements such as promoters are finely regulated by overall environmental or culture conditions . In such a framework, relative measurements of promoter activities may be both easier to obtain and more relevant. Many natural biological systems already follow this model, and are robust to the absolute properties of components so long as the relative relationships between subparts are maintained. For example, developmental body plans may vary in size with individual organisms having different overall sizes, however the ratio of the sizes of individual bones or organs to overall body mass is often tightly maintained .
Standard promoter definition
The promoters tested here were practically standardized to have identical transcription initiation sites (predicted) and identical sequences downstream of the initiation site (Additional file 1). Thus, we expect that the mRNA expressed by each of the tested promoters is identical to the mRNA produced by the reference standard promoter, and that we can cancel the mRNA degradation rate and translation rate of immature GFP from mRNA terms in simplifying the model relating GFP synthesis rates to RPUs. This simplification allows the activity of promoters to be reported in comparable units (RPUs) without needing to directly measure mRNA levels. Going forward, an expanded definition for standard BioBrick promoters could be developed in order to ensure that all promoters share the same transcription start position and a fixed sequence downstream of the transcription start site (that is, 5' mRNA UTR). All promoters that adhered to such a standard could then be reliably measured using the kits described here, or via future kits based on gene expression reporters that adhered to any new standard, without the need for promoter-specific mRNA quantitation.
Distribution, use, and improvement of standardized measurement kits
Shared measurement tools and reference standards become more useful as they are broadly adopted. To facilitate such adoption, the Registry of Standard Biological Parts now includes our promoter measurement kit and reference standard in the annual distribution of BioBrick parts. We also created a website in order to support the reporting and sharing of promoter activity measurements http://partsregistry.org/measurement. This website contains instructions for use of the kit and summarizes previously characterized promoters. Finally, to enable discussion of proposed improvements to the kit and reference standard, and also the development of new kits and reference standards, we are supporting an open discussion of technical standards in synthetic biology http://biobricks.org/standards.
Standard tools, techniques, and units for measurement are needed for a distributed community of biological engineers to independently characterize and share biological parts. We have defined a shared unit for measuring relative promoter activity (Reference Promoter Units, RPUs) and demonstrated that relative promoter activity can address some of the challenges in measurement across labs due to varying experimental conditions and measurement instruments. We developed a first-generation measurement kit for BioBrick promoters, and are freely distributing the kit via the Registry of Standard Biological Parts. Having demonstrated the feasibility and ease of use of the kit, we hope to encourage a community of users to adopt and improve these measurement tools and reference standard in order to characterize promoters via a comparable and common unit, the RPU. We expect that the shared experiences of biological engineers using common measurement tools and standards will help to identify new engineering challenges in improving the reliability and reuse of standard biological parts.
Strains and media
All measurement experiments and cloning were performed in E. coli TOP10 (Invitrogen) or W3110. Supplemented M9 minimal medium (M9 salts, 1 mM thiamine hydrochloride, 0.2% casamino acids, 0.1 M MgSO4, 0.5 M CaCl2) was used for all measurement experiments with either glycerol (0.4%) or glucose (0.4%) added as a carbon source and kanamycin (20 μ g/ml) antibiotic added where appropriate. All oligonucleotides were purchased from Invitrogen and DNA modifying enzymes were purchased from New England Biolabs.
Promoter measurement kit contents
Sequences for all BioBrick plasmids (denoted pSB***) and BioBrick parts (denoted BBa_####) are available through the Registry of Standard Biological Parts . pSB3K3 contains a p15A origin of replication (copy number 10–12) and the kanamycin resistance marker, pSB4T5 contains the pSC101 origin of replication (copy number ~5) and the tetracycline resistance marker, and pSB3C5 contains the p15A origin of replication and the chloramphenicol resistance marker. Physical copies of the plasmids and parts are also available from the Registry via the annual Registry parts distribution. The details of the promoter measurement kit contents are described in Supplementary Box 1 and Supplementary Table 1 (see Additional file 1). The sequences for the preparative primers used to amplify pSB3K3 to generate backbone plasmid are: TACTAGTAGCGGCCGCTGCAG (forward primer) and CTCTAGAAGCGGCCGCGAATTC (reverse primer).
Assembly of test constructs
We built promoters by annealing synthesized oligonucleotides. The oligonucleotides were ordered with 5' phosphates and designed to leave an EcoRI overhang on the 5' end and a SpeI overhand on the 3' end so they could be used in subsequent ligation reactions without an intermediate restriction digest step. We inserted seven promoters: BBa_J23113, BBa_J23116, BBa_J23150, BBa_J23151, BBa_J23102, BBa_R0040, and BBa_R0011 into the promoter test construct and transformed into TOP10 according to the process outlined in Supplementary Box 1 (see Additional file 1). We found the optimal concentration of DNA for each of the three components in the ligation reaction (pSB3K3, BBa_E0240 or BBa_I13401, and the test promoter) was approximately 10 ng per uL. More detailed protocols and troubleshooting can be found at http://partsregistry.org/measurement. In the process of construction we found mutations in two of the promoters that we attribute to errors in the synthesis of the oligonucleotides that were annealed to construct the promoters. The two promoters were functional so we included them as additional members of the collection (BBa_J23150 and BBa_J23151). The method of part assembly described here is based on the three-antibiotic BioBrick standard assembly method .
Assay of promoter collection
The protocol described here will be referred to as the "original" protocol throughout the methods section and describes the measurement procedure used to characterize the set of seven promoters (Figure 3). For each promoter construct three 17 mm test tubes containing 5 ml of pre-warmed (37°C) supplemented M9 medium with kanamycin (20 μ g/ml) were inoculated from single colonies of TOP10-DH5α containing the promoter test construct on the pSB3K3 vector backbone. Cultures were grown in 17 mm test tubes for approximately 20 hrs at 37°C with spinning at 70 rpm. We then diluted the cultures 1:100 into 5 ml of pre-warmed fresh media and the cultures were grown for approximately four hours under the previous conditions (17 mm tubes, 37°C, spinning at 70 rpm). After four hours, we measured the OD600 of a 500 μ l aliquot from each culture on a WPA Biowave Spectrophotometer. Based on this OD measurement, the cultures were diluted to the same OD (0.07) in 5 ml of pre-warmed fresh media and grown for one hour at 37°C. We then transferred three 200 μ l aliquots from each culture into a flat-bottomed 96 well plate (Cellstar Uclear bottom, Greiner). We incubated the plate in a Wallac Victor3 multi-well fluorimeter (Perkin Elmer) at 37°C and assayed with an automatically repeating protocol of absorbance measurements (600 nm absorbance filter, 0.1 second counting time through 5 mm of fluid), fluorescence measurements (485 nm excitation filter, 525 nm emission filter, 0.1 seconds, CW lamp energy 12901 units), and shaking (3 mm, linear, normal speed, 15 seconds).
Since we are calculating a ratio of the GFP synthesis rates we do not need to determine each rate in absolute units of GFP per second per cell, rather we can use the background-subtracted fluorescence (F) that is proportional to the number of GFP molecules and the background-subtracted absorbance (ABS) that is proportional to the number of cells in the culture to calculate the ratio of GFP synthesis rates [9, 36].
Assay of different measurement conditions
We measured the promoter activity of two promoters (BBa_J23101 and BBa_J23150) under seven different measurement procedures. The first of the seven procedures was identical to the "original" protocol described above for measuring the 7-member promoter collection except it was conducted at 30°C in the strain W3110 with pSB3K3 as the vector backbone for the promoter test construct. The second procedure was identical to the original except it was conducted at 30°C. The third procedure was identical to the original except that it was conducted at 30°C and pSB4T5 was used as the vector backbone. The fourth procedure was identical to the original except that it was conducted at 30°C and used pSB3C5 as the vector backbone. The fifth procedure was identical to the original. The sixth procedure was identical to the original except that instead of the second dilution into tubes followed by 1 hour of growth, the cells were diluted into 96 well plates and incubated for two hours before we started taking measurements. The seventh procedure was identical to the sixth except glucose was used instead of glycerol as the carbon source.
Assay of inter-laboratory variability
We distributed a set of four promoters (BBa_J23113, BBa_J23150, BBa_J23151, and BBa_J23102) to six laboratories to take independent measurements of promoter activity. The protocol each lab conducted was identical to the original protocol described, except that the cells were harvested after the first 1:100 dilution and 4 hours of growth (there was no second dilution step). The cells were then spun down, resuspended in PBS, and the fluorescence per cell was measured using a flow cytometer. The measurement equipment used (cytometer model, laser, emission filter) varied between the laboratories (see Additional file 1 [Supplementary Table 2]).
μ φ /μJ 23101is a correction term based on differences in growth rate. Changes in the growth rate effect per cell GFP accumulation since loss of GFP per cell is largely due to dilution. Since we are calculating a ratio of the per cell GFP concentrations we do not need to determine each rate in absolute units of GFP molecules per cell, rather we can use the background-subtracted per cell fluorescence ([F]) that is proportional to the number of GFP molecules per cell to calculate the ratio of GFP concentrations.
After background correction, the per cell fluorescence ([F]) was determined for each promoter and activities in RPUs were calculated using Eq. 10. We applied the growth rates measured previously (Additional file 1 [Supplementary Table 3]) across all laboratories when calculating RPUs, rather than requesting individual laboratories to measure growth rates. This approximation likely increased the variability in the promoter activity measurements across laboratories, as growth rates vary between laboratories due to differences in culture conditions and media.
The authors would like to thank Nishant Baht and Karen Wong for building and testing alternate designs of the measurement kits. The authors would like to thank Barry Canton for helpful discussions. The authors would like to thank Julius B. Lucks, Lei Qi, Weston Whitaker and Gavin Price for taking measurements in the multi-laboratory study. JC and AG would like to acknowledge: Stephanie Terrizzi and the Flow Cytometry and Cell Sorter Facility for flow cytometry (FACSAria funded by NCCR SIG grant # 1S10RR021051-01A2). CA would like to recognize that the work was performed in part at the Molecular Foundry, Lawrence Berkeley National Laboratory, with support from the Office of Science, Office of Basic Energy Sciences, of the U.S. Department of Energy under Contract No. DE-AC02-05CH11231. Funding for this work was provided by the NSF Synthetic Biology Engineering Research Center. JK received additional support via an NSF Graduate Research Fellowship.
- Knight T: Idempotent Vector Design for Standard Assembly of Biobricks.2003. [http://web.mit.edu/synbio/release/docs/biobricks.pdf]Google Scholar
- Voigt CA: Genetic parts to program bacteria. Current Opinion in Biotechnology 2006, 17: 548-557. 10.1016/j.copbio.2006.09.001View ArticleGoogle Scholar
- Arkin AP, Endy D: A Standard Parts List for Biological Circuitry. DARPA White Paper 1999. [http://hdl.handle.net/1721.1/29794]Google Scholar
- Endy D: Foundations for engineering biology. Nature 2005, 438: 449-453. 10.1038/nature04342View ArticleGoogle Scholar
- Endy D: 2003 Synthetic Biology study.2003. [http://hdl.handle.net/1721.1/38455]Google Scholar
- Knight T: DARPA BioComp Plasmid Distribution 1.00 of Standard Biobrick Components.2002. [http://hdl.handle.net/1721.1/21167]Google Scholar
- Texas Instruments: TTL logic data book: standard TTL, Schottky, low-power Schottky. Texas Instruments, Dallas, Texas, USA; 1988.Google Scholar
- Registry of Standard Biological Parts[http://partsregistry.org]
- Canton B, Labno A, Endy D: Refinement and standardization of synthetic biological parts and devices. Nature biotechnology 2008, 26: 787-793. 10.1038/nbt1413View ArticleGoogle Scholar
- Rosenfeld N, Young JW, Alon U, Swain PS, Elowitz MB: Accurate prediction of gene feedback circuit behavior from component properties. Mol Syst Biol 2007, 3: 143. 10.1038/msb4100185View ArticleGoogle Scholar
- Ajo-Franklin CM, Drubin DA, Eskin JA, Gee EPS, Landgraf D, Phillips I, Silver PA: Rational design of memory in eukaryotic cells. Genes Dev 2007, 21: 2271-2276. 10.1101/gad.1586107View ArticleGoogle Scholar
- Guido NJ, Wang X, Adalsteinsson D, McMillen D, Hasty J, Cantor CR, Elston TC, Collins JJ: A bottom-up approach to gene regulation. Nature 2006, 439: 856-860. 10.1038/nature04473View ArticleGoogle Scholar
- Serebriiskii IG, Golemis EA: Uses of lacZ to Study Gene Function: Evaluation of [beta]-Galactosidase Assays Employed in the Yeast Two-Hybrid System. Analytical Biochemistry 2000, 285: 1-15. 10.1006/abio.2000.4672View ArticleGoogle Scholar
- Miller JH: Experiments in Molecular Genetics. Cold Spring Harbor Laboratory, USA; 1972.Google Scholar
- Eustice DC, Feldman PA, Colberg-Poley AM, Buckery RM, Neubauer RH: A sensitive method for the detection of beta-galactosidase in transfected mammalian cells. BioTechniques 1991, 11: 739-740. 742-733-739-740, 742-733Google Scholar
- Pollack A: Custom-Made Microbes at Your Service. New York Times. New York City 2006.Google Scholar
- Coefficient of thermal expansion[http://en.wikipedia.org/wiki/Coefficient_of_thermal_expansion]
- Barnett E, Nordin BE: The radiological diagnosis of osteoporosis: a new approach. Clinical Radiology 1960, 11: 166-174. 10.1016/S0009-9260(60)80012-8View ArticleGoogle Scholar
- Schena M, Shalon D, Davis RW, Brown PO: Quantitative monitoring of gene expression patterns with a complementary DNA microarray. Science (New York, NY) 1995, 270: 467-470.View ArticleGoogle Scholar
- Alper H, Fischer C, Nevoigt E, Stephanopoulos G: Tuning genetic control through promoter engineering. Proceedings of the National Academy of Sciences of the United States of America 2005, 102: 12678-83. 10.1073/pnas.0504604102View ArticleGoogle Scholar
- Weiss R: Cellular computation and communications using engineered genetic regulatory networks.Massachusetts Institute of Technology; 2001. [http://hdl.handle.net/1721.1/8228]Google Scholar
- Endy : Adventures in synthetic biology. Nature 2005, 438: 449-453. 10.1038/nature04342View ArticleGoogle Scholar
- Leveau JHJ, Lindow SE: Predictive and Interpretive Simulation of Green Fluorescent Protein Expression in Reporter Bacteria. Journal of Bacteriology 2001, 183: 6752-62. 10.1128/JB.183.23.6752-6762.2001View ArticleGoogle Scholar
- Adams CW, Hatfield GW: Effects of promoter strengths and growth conditions on copy number of transcription-fusion vectors. J Biol Chem 1984, 259: 7399-7403.Google Scholar
- Shetty R, Endy D, Knight T: Engineering BioBrick vectors from BioBrick parts. Journal of Biological Engineering 2008, 2: 5-5. 10.1186/1754-1611-2-5View ArticleGoogle Scholar
- Hawley DK, McClure WR: Compilation and analysis of Escherichia coli promoter DNA sequences. Nucleic acids research 1983, 11: 2237-2255. 10.1093/nar/11.8.2237View ArticleGoogle Scholar
- Bernstein JA, Khodursky AB, Lin P-H, Lin-Chao S, Cohen SN: Global analysis of mRNA decay and abundance in Escherichia coli at single-gene resolution using two-color fluorescent DNA microarrays. Proceedings of the National Academy of Sciences 2002, 99: 9697-9702. 10.1073/pnas.112318199View ArticleGoogle Scholar
- OpenWetWare: FACS protocol for multi-institution promoter activity experiment[http://openwetware.org/index.php?title=The_BioBricks_Foundation:Standards/Technical/Measurement/Promoter_characterization_experiment_%28FACS%29&oldid=198396]
- Lutz R, Bujard H: Independent and tight regulation of transcriptional units in Escherichia coli via the LacR/O, the TetR/O and AraC/I1-I2 regulatory elements. Nucleic acids research 1997, 25: 1203-1210. 10.1093/nar/25.6.1203View ArticleGoogle Scholar
- Basu S, Gerchman Y, Collins CH, Arnold FH, Weiss R: A synthetic multicellular system for programmed pattern formation. Nature 2005, 434: 1130-1134. 10.1038/nature03461View ArticleGoogle Scholar
- International Genetically Engineered Machines (iGEM) Competition[http://igem.org]
- Bremer H, Dennis P: Modulation of chemical composition and other parameters of the cell by growth rate. In Escherichia coli and Salmonella. Volume 2. 2nd edition. Edited by: Neidhart FC. Washington DC: ASM Press; 1996:1553-1569.Google Scholar
- Huxley J: Problems of Relative Growth. Methuen, London; 1932.Google Scholar
- Kalir S, McClure J, Pabbaraju K, Southward C, Ronen M, Leibler S, Surette MG, Alon U: Ordering Genes in a Flagella Pathway by Analysis of Expression Kinetics from Living Bacteria. Science 2001, 292: 2080-2083. 10.1126/science.1058758View ArticleGoogle Scholar
- Setty Y, Mayo AE, Surette MG, Alon U: Detailed map of a cis-regulatory input function. Proceedings of the National Academy of Sciences 2003, 100: 7702-7707. 10.1073/pnas.1230759100View ArticleGoogle Scholar
- Ronen M, Rosenberg R, Shraiman BI, Alon U: Assigning numbers to the arrows: Parameterizing a gene regulation network by using accurate expression kinetics. Proceedings of the National Academy of Sciences 2002, 99: 10555-10560. 10.1073/pnas.152046799View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.